This machine is no longer manufactured and rarely installed. When found in the secondary market, most systems are often used for parts.
Accuray Cyberknife G4 Linear Accelerator
manufacturer: Accuray model: Cyberknife G4reference number: 17H23year: 2007availability: Available
- Manufactured in 2007
- Max Dose Rate: 600 mu/min
- Software Version is 9.6.02
- Standard Couch
- Thyratron and DQing thyratron both changed in September 2019.
- System does 5-8 patients a week, since install.
- Magnetron changed in April 2019.
- Pulse shaper changed in April 2019.
- Water pump and motor changed in August 2019.
- Laser pointer for system, power supply changed in May 2019.
- System has been maintained by Accuray and Hospital Biomed since install.
- System has been under a first look contract since install.
- Cabling is 4 years old
- Robot has been maintained by Kuka (The Robot manufacturer) since install.
- Wall lasers included, power conditioner Included.
- KV generators and detectors are original.
- Linac is Original, it has not been replaced (Approx. 13.5 years old)
- Last patient was treated January 24, 2020
- Currently Installed and Operational, but not in clinical use
Please send me more information regarding this Accuray Cyberknife G4 Linear Accelerator.